ATBECHDEDKESFRGBGRITLILUNLSEMCPTIE......FI....CY....................................................
*
J
JDIM360 Ver 1.28 (29 Oct 2014) - 2999001/0
1214582
CORRECTED EUROPEAN PATENT SPECIFICATION
B9
B1
20150218
W1
de
Zeichnung(en) ersetzt oder hinzugefügt
en
Drawing(s) replaced or added
fr
Figure(s) remplacée(s) ou ajoutée(s)
de
Zeichnungen
en
Drawings
fr
Figures
EP
00965009.4
20000916
20020405
20070227
en
en
en
398129
19990917
US
20150218
201508
20020619
200225
20140716
201429
20140210
20150218
201508
G01N 5/04 20060101AFI20140121BHEP
G01N 1/44 20060101ALI20140121BHEP
G01N 1/40 20060101ALI20140121BHEP
G01N 22/00 20060101ALI20140121BHEP
G01N 22/04 20060101ALI20140121BHEP
de
MIKROWELLENANALYSATOR FÜR FLÜCHTIGE STOFFE MIT HOCHEFFIZIENZHOHLRAUMRESONATOR
en
MICROWAVE VOLATILES ANALYZER WITH HIGH EFFICIENCY CAVITY
fr
ANALYSEUR HYPERFREQUENCES DE MATIERES VOLATILES COMPORTANT UNE CAVITE A RENDEMENT ELEVE
WO-A-99/40409
US-A- 3 890 825
US-A- 4 908 486
US-A- 5 632 921
VARLAMOVA N A ET AL: "ON THE QUALITY FACTOR OF CAVITY RESONATORS OF DIFFERENT SHAPES" TELECOMMUNICATIONS AND RADIO ENGINEERING,US,BEGELL HOUSE, INC., NEW YORK, NY, vol. 52, no. 2, 1998, pages 79-85, XP000834425 ISSN: 0040-2508
JENNINGS, William, Edward
5722 Olive Branch Road
Wingate, NC 28174
US
BARRETT, Matthew, Donald
1200 St. Andrews Road, Apt. 901
Columbia, SC 29210
US
KING, Edward, Earl
4709 Pineland Place
Charlotte, NC 28277
US
CEM Corporation
100096866
SEM/CH/36894
P.O. Box 200 3100 Smith Farm Road
Matthews, NC 28106-0200
US
Murgatroyd, Susan Elizabeth
et al
100026841
Baron Warren Redfern Cambridge House 100 Cambridge Grove
Hammersmith London W6 0LE
GB
AT
BE
CH
CY
DE
DK
ES
FI
FR
GB
GR
IE
IT
LI
LU
MC
NL
PT
SE
US2000025199
20000916
en
WO2001020312
20010322
200112

FIELD OF THE INVENTION
The present invention relates to analytical chemistry techniques and instruments, and particularly those used for analyzing the volatiles content of materials. In particular, the invention relates to the use of microwave energy to drive moisture or other volatiles from substances so that the moisture content (or in complimentary fashion the solids content) can be quickly and accurately measured.

BACKGROUND OF THE INVENTION
The analysis of the volatiles content of materials is one of the most common types of testing carried out on a wide variety of materials. In its most fundamental sense, the measurement of the volatile (very often the moisture) content of a material is usually carried out by taking a representative sample of the material, weighing it, drying it, and then re-weighing it after the drying process. The difference between the starting weight and the ending weight, when divided by the starting weight and expressed as a percentage, is the percentage of volatile content. The procedure is based on differential weighing and is somewhat formally referred to as gravimetric moisture determination.
As a preliminary matter, it is well-understood in this art that the results of such differential weight analysis can be calculated and expressed either as the percentage of volatiles that leave a sample, or as the percentage of solids that remain. The manipulative steps, however, are the same. Thus, devices of the type described herein are often described as "moisture/solids" analyzers.
Furthermore, although the term "moisture" is used most frequently herein, it will be understood that the apparatus and methods disclosed and claimed herein apply to any volatile species that can be driven from a heated sample in a differential weighing technique.
For many years, moisture analysis of this type has been carried out by heating the respective samples in conventional conduction or convection-type ovens. The process is generally time consuming for several reasons. First, in order to make sure that all of the moisture has been driven from a sample, the heating and re-weighing process must be carried out several times until the difference in dry weights are either eliminated, or so small as to be within measurement accuracy limits. Because conventional drying is relatively time consuming, the need to take several repeated measurements on the same sample is similarly time consuming.
As another factor, the analytical balances typically used to weigh the samples are usually pan-type balances; i.e., they include a flat surface upon which the sample (and sometimes its container) are placed and which is in turn attached to a sensitive balancing mechanism. Under these circumstances, when a warm object such as a heated sample is placed on a balance pan, the warm object tends to set up a flow of convection air currents in the immediate vicinity. In turn, these upwardly flowing currents tend to lift the balance pan and produce an inaccurate reading. Generally, the more sensitive the balance, the more likely, or the greater amount proportionally by which, the reading will be in error. Thus, in addition to the time required to conventionally heat a sample to dryness on a repeated basis, there also exists the need to allow the sample to cool sufficiently to avoid creating convection currents in the balance. Thus, convection and conduction methods of moisture analysis tend to be relatively time consuming.
More recently, microwave energy has been used to help speed up the drying process. In these techniques, microwaves are used to drive the moisture from the sample rather than conventional convection or conduction heating. Microwaves offer several advantages in this respect, the most direct being the fact that they heat water or other volatiles directly, rather than by conduction through the material itself. Stated differently, microwaves immediately interact with the moisture in a sample and tend to volatize it quickly. Furthermore, because microwaves affect only certain types of substances within a sample (generally polar substances), they tend to heat the overall sample less than do conventional convection and conduction techniques. As a result, microwave heating can hasten moisture analysis by several orders of magnitude. By way of example, different manufacturers of microwave drying devices point out that processes that can take as many as 16 hours for moisture analysis (e.g., ground beef) can be done in a conventional microwave device in about 5 minutes. Additionally, even those materials which can be dried relatively quickly in conventional ovens, can still be greatly accelerated in a microwave device. For example, according to published information, tomato paste (which has a moisture level of about 77.55%) can be dried in a conventional oven in about 1.5 hours. In contrast, it can be dried in a conventional microwave device in about 5 minutes. Other materials, such as cheese, can be analyzed in a conventional microwave drying device in as little as 3.5 minutes.
Accordingly, the microwave moisture analyzer has become a widely accepted piece of apparatus in many chemical laboratories. An exemplary version is the Lab Wave-9000 microwave moisture/solids analyzer from CEM Corporation, the assignee of the present invention, which combines an analytical balance with its microwave drying capability. Devices such as the Lab Wave-9000 are also typically operated in conjunction with microprocessors and related electronic circuits. The processor's operation and logic (software) enable the moisture content to be calculated even more quickly using generally well understood relationships between weight loss over a given period of time and the ultimate moisture content of a given sample.
Nevertheless, the wide acceptance of microwave drying devices such as the Lab Wave-9000 has driven further innovation and a desire to carry out drying processes even more quickly and efficiently. Accordingly, newer types of microwave devices are appearing in the marketplace which attempt to focus microwave energy more carefully and efficiently in an attempt to dry materials more quickly, more accurately, and using less power.
Some newer versions of microwave drying devices tend to follow the disclosures of U.S. Patent No. 5,632,921 to Per O. Risman. The Risman patent describes a generally cylindrical microwave cavity device in which the modes of microwave energy are carefully controlled to give a heating effect. Based on public disclosures, and without attempting to speak for the respective manufacturers, the "Moistwave" system from Prolabo (France) and the "M2" microwave moisture/solids analyzer from Denver Instruments (Arvada, Colorado, USA) appear to follow the Risman patent in their use of specifically cylindrical cavities. According to advertising materials for both of these devices, they are able to reduce even the ordinary microwave drying time by a significant fraction, for example taking drying steps that require 3, 4, or 5 minutes in a conventional microwave device, and carrying them out in about 1 or 2 minutes or less. As set forth in the Risman '921 patent, however, these devices operate on a very carefully selected combination of limited transverse magnetic modes.
The Denver Instruments device also appears to follow some or all of the disclosures of published International Application WO 99/40409 corresponding to International Application PCT/US/01866.
Accordingly, a need exists for further improvement in the area of high-efficiency microwave moisture analyzers.
Therefore it is an object of the present invention to provide a microwave volatiles (or "moisture/solids") analyzer that is efficient and thorough in its analysis, while minimizing the amount of energy and physical space that it uses.
The present invention provides a volatiles analyzer comprising:
- a source of microwave radiation that can selectively produce at least one predetermined frequency of microwave radiation;
- a cavity in communication with said source; and
- an analytical pan balance with at least its balance pan in said cavity;
- and wherein the walls of said cavity form a polyhedron that focuses microwave energy of the predetermined frequency on said balance pan while supporting a plurality of TM and TE modes in said cavity,
- characterized in that said cavity comprises a polyhedron with twelve faces, eight said faces having planes which form a regular octagon;
- wherein one said face comprises a regular octagon face perpendicular to said eight faces;
- wherein a further face of said polyhedron is an octagon parallel to said regular octagon face; and
- wherein two of said faces are both non-parallel and non-perpendicular to the other ten said faces.
Preferred embodiments of the invention will now be described with reference to the detailed description and with the accompanying drawings in which:
- FIG. 1 is a perspective view of a commercial embodiment of a volatiles analyzer according to the present invention;
- FIG. 2 is another perspective view of the volatiles analyzer of the present invention and showing it in the open position along with portions of the interior of the cavity;
- Figure 3 is a perspective view of the volatiles analyzer of the present invention without the cover and showing some of the internal parts;
- Figure 4 is a view similar to Figure 3 but taken from the opposite side of the volatiles analyzer;
- Figure 5 is a perspective view taken into the upper portions of the cavity of the present invention;
- Figure 6 is a top cross-sectional view taken along lines 6-6 of Figure 3;
- Figure 7 is a front elevational view of the cavity according to the present invention;
- Figure 8 is a cross-sectional view of the cavity taken along lines 8-8 of Figure 5;
- Figure 9 is a bottom plan view of the top half of the lid and cavity of the present invention; and
- Figure 10 is a perspective view of the air shield portion of the present invention.
Figure 1 is a perspective view of a commercial embodiment of the present invention broadly designated at 10. The microwave portions of the device are in most circumstances kept under a cover 11 and will be described in more detail with respect to the other figures. The cover 11 is secured to a base 12 by a latch 13. Figure 1 also shows the housing 14 for any microprocessors used in conjunction with the device, along with a display 15 and a keyboard entry pad 16. In the embodiment shown in Figure 1, the indented portion 17 near the rear of the device 10 is shaped to hold a number of sample pads of the type that are typically used in this type of moisture analysis.
Figure 2 shows the volatiles analyzer 10 of the present invention in an open position. Figure 2 illustrates that the invention includes an analytical pan balance of which at least the balance pan 20 is present in a cavity 21. It will be understood that the term "balance pan" is used in its broadest sense and does not literally have to be a pan. Instead, as illustrated in Figure 2 and elsewhere, the pan is preferably formed of a framework, often of a material such as plastic, that is substantially permeable or even transparent to microwave radiation so as not to otherwise interfere with the processes of the device. In a typical operation, a sample pad (not shown) is placed on top of the pan 20 and a sample is in turn placed on top of the sample pad. For some materials, a second sample pad is placed over the sample to prevent splattering during the heating process. The preferred sample pads are preferably formed of glass fibers, although any material is acceptable that does not otherwise interfere with the microwaves or other aspects of the process.
The nature and operation of analytical balances are generally well understood in the chemical arts and will not be described in detail herein. The balance should be selected so as to avoid interfering with the propagation of microwaves in the cavity in any undesired fashion. Various balances are available from, or can be readily custom-designed by, manufacturers such as Mettler Toledo (Mettler Toledo International, Inc., Worthington, Ohio, USA). Some such balances can measure with an accuracy approaching 0.0000001 gram (g). For most moisture determinations on small samples an accuracy of 0.0001 g (i.e., 0.1 milligram) is preferred.
Figure 2 also shows the cavity broadly designated at 21 which in the invention forms a polyhedron that focuses microwave energy of the predetermined frequency generated by the source (Figure 3) on the balance pan 20 while supporting a plurality of TM and TE modes in the cavity. By focusing the energy on the balance pan-i.e., on the pan's geometric position in the cavity-the invention greatly enhances the efficiency of the heating process, even though the cavity itself is much smaller than in earlier devices.
Figure 2 also shows the port 22 that permits entry of microwave energy from the waveguide (Figure 3) into the cavity 21. In preferred embodiments, the port 22 is the sole entry for microwaves between the waveguide and the cavity 21. As perhaps best understood by a combination of Figures 2 and 3, in preferred embodiments the waveguide 18 is a rectangular solid and the port 22 is a longitudinal slot positioned in one face of the waveguide 18 and oriented neither parallel nor perpendicular to any of the angles forming the rectangular solid waveguide.
Figure 2 also illustrates that the invention can be understood in another sense as a cavity for a microwave device that comprises a polyhedron in which eight of its faces form a regular octagon. The words "polyhedron," "octagon," and "regular" are all used in their ordinary and dictionary sense in this specification and in the claims. Thus, a polyhedron is a three-dimensional geometrical solid with a plurality of planar faces. An octagon is a two-dimensional figure with eight linear sides. A regular octagon is an eight-sided two-dimensional figure in which each of the eight sides are equal in length and all of the eight angles are equal to one another.
Figure 2 also shows that in the illustrated embodiment, one face is a regular octagon illustrated as 23 and comprises the bottom face of the cavity 21. Figure 2 also illustrates that in the preferred embodiment, the polyhedron has 12 sides, eight of which (19, 24, 25, 29-31, and 38-39) are both joined to and perpendicular to the regular octagon face 23. As perhaps best illustrated in Figure 3, the polyhedron cavity also includes two faces, 26 and 28 that are both nonparallel and nonperpendicular to the other ten faces of the 12-sided polygon. The parallel face 27 (i.e., parallel to the regular octagon face) is joined to two of the perpendicular faces that are respectively parallel to one another, the nonparallel and nonperpendicular faces are each joined to the parallel face 26, and the nonparallel and nonperpendicular faces are each joined to the perpendicular faces. As Figure 2 illustrates, the eight sides 19, 24, 25, 29-31, and 38-39 form a regular octagon regardless of the top and bottom geometry of the cavity 21.
Although the illustrated embodiment shows the bottom face 23 as being planar and a regular octagon, the invention is not limited to such an arrangement, and the bottom portion of the cavity could be nonplanar. Those familiar with the propagation of microwave energy will recognize that the modes and focusing of the microwaves depend on the waveguide, the port from the waveguide to the cavity, and the cavity geometry, and not upon visual references or orientations such as "up" or "down" or the "top," "bottom," or "sides" of the cavity. Such terms are used herein in conjunction with the drawings to illustrate the invention rather than to limit its scope.
Figure 2 also shows a number of items that are the same as and were described with respect to Figure 1, and that will not be otherwise described with respect to Figure 2.
Figure 3 illustrates the source and waveguide as well as other aspects of the invention that have already been referred to. The source of microwave radiation typically comprises a magnetron illustrated at 32 in Figure 3. The source can, however, comprise a klystron or a solid-state device. As noted earlier, the source 32 is in communication with the waveguide 23 which in turn is in communication with the cavity 21. Figure 3 also illustrates a cooling system 33 which takes the form of a pan (Figure 4).
Figure 3 also illustrates that in preferred embodiments, the moisture analyzer includes an infrared sensor 34 positioned with respect to the cavity 21 to measure the temperature of a sample on the balance pan 20. The infrared sensor 43 is in electronic communication with means for moderating the microwave power in the cavity based upon the measured temperature. The moderation of temperature can be as simple as temporarily turning off the power from the magnetron 32, or it can be a more sophisticated control using, for example, a switching power supply. The basic use of the infrared sensor, and its advantages for processing certain types of items, are set forth in more detail in commonly assigned International application No. PCT/US99/21490. The use of the switching power supply to control microwave energy is likewise set forth in commonly assigned U.S. Patent No. 6,084,226.
The moisture analyzer of the present invention typically further comprises a processor (not shown) in electronic communication with the balance for calculating the expected final weight of a sample based upon two or more measurements of the weight at discrete time intervals as the weight changes under the influence of microwave radiation upon the sample. The algorithms for calculating such final weight based on the weight change are generally well understood in this art and will not be set forth in detail herein. An early version is set forth, by way of example and not limitation, in U.S. Patents Nos. 4,438,500 and 4,457,632, which are commonly assigned with the present application. Indeed, the use of particular algorithms and processor logic are often the choice of individual designers and can be carried out by those of ordinary skill in this art without undue experimentation. As another exemplary source, Dorf, The Electrical Engineering Handbook, 2d Ed. (CRC Press 1997), contains extensive discussions of electronic controls, circuits, and processor logic.
One of the advantageous features of the invention is its capacity of the cavity for supporting a plurality of TM and TE modes at the wavelengths produced by the source. As known to those of ordinary skill in this art, a mode is one of the various possible patterns of propagating or standing electromagnetic fields in a particular waveguide or cavity. A mode is characterized by its frequency, polarization, electric field strength, and magnetic field strength. The distribution of modes in a cavity will always satisfy Maxwell's equations and can be calculated to a fairly accurate extent, particularly using the available computational capacity of modem microprocessors. In the same manner, the term "field" is used in its usual sense to indicate the volume of influence of a physical phenomenon expressed as vectors. TM and TE modes refer to transverse electric and transverse magnetic modes. A transverse mode is one whose field vectors, whether electric or magnetic, are both normal to the direction of propagation of the wave energy.
The remainder of the figures illustrate these and other features of the invention. Figure 4 illustrates the power supply 34 for the magnetron 32 as well as a clearer illustration of the pan 33. Figure 4 also illustrates the power switch 35 which is also referred to as a power entry module. A series of sub-D connectors 36, 37, and 40, provide for optional communication with stand-alone printers or computers or similar devices. The sub-D connectors 36, 37, and 40 provide parallel and serial connections in a manner well understood by those of ordinary skill in these arts and the nature of the connections will not be otherwise explained in detail. Figure 4 also illustrates the printer portion 41 of the device which, if desired, can take the results from any particular test, and print them out all on a paper tape. As just noted, the results can likewise be forwarded to digital memory, a digital processor, or any other location using the sub-D connectors. Figure 4 also illustrates a number of the items which have already been described, and which carry corresponding numerals in Figure 4, but which will not be otherwise described in detail.
Figure 5 is a perspective view of the cover portion 11 of the analyzer 10, and showing a view looking upwards into the cavity 21. Figure 5 illustrates the slot 22, the polyhedron shape of the cavity, and also illustrates the air shield 42 that is used in preferred embodiments of the invention. The nature, structure, function, and operation of the air shield 42 are set forth in greater detail in co-pending and commonly assigned U.S. application Serial No. 09/397,825, filed September 17, 1999, for "Microwave Apparatus and Method for Achieving Accurate Weight Measurements."
Figure 6 is a top plan view of the cavity of the present invention and particularly illustrates the regular octagon shape of the bottom place 23 of the cavity, which in turn defines the cross-sectional shape of the cavity as taken from this view. Figure 6 also shows the waveguide 18 adjacent to the cavity 21.
Figure 7 is a front elevational view of the cavity 21, and shows the perpendicular sides 19, 24, 25, 29-31, 38, and 39, as well as one of the nonparallel sides 26, which, in preferred embodiments of the invention, is perforated to allow for an airflow through the cavity 21 while the moisture analysis is taking place.
Figure 8 is a cross-sectional view of the cavity 21 and showing the cross-sectional profile of the air shield 42 in one of its preferred positions.
Figure 9 is a bottom plan view of the cover portion 11, the cavity 21, and the air shield 42. Figure 9 illustrates that both of the nonparallel, nonperpendicular sides are perforated to encourage the desired airflow, or airflow that can be enhanced using the fan 33.
Figure 10 illustrates a preferred embodiment of the air shield 42 which is generally formed of a grid-like framework 44 which carries either a single, or a plurality of moisture absorbent pads 45. As set forth in the co-pending application, these pads absorb water vapor driven from a heated sample and allow it to recondense. By first absorbing the vapor, the shield 42 moderates the flow of gases through the cavity that could otherwise disturb the accuracy of the balance pan. Furthermore, as is known to those familiar with microwave techniques and characteristics, water vapor is affected differently by microwaves than liquid water and thus the absorbent pads provide a location where the water vapor can recondense, again be heated by the microwaves, and then be carried off by the fan. As set forth in the co-pending application, the fan preferably draws an airflow across the upper portions of the cavity 21 generally perpendicular to the expected flow of rising water vapor from the sample. In this manner, the combination of the airflow and the shield help draw off the water vapor in a manner that disturbs the balance pan the least, and preferably not at all.
In the drawings and specification, there have been disclosed typical embodiments of the invention, and, although specific terms have been employed, they have been used in a generic and descriptive sense only and not for purposes of limitation, the scope of the invention being set forth in the following claims.
1. A volatiles analyzer (10) comprising:
a source of microwave radiation that can selectively produce at least one predetermined frequency of microwave radiation;
a cavity (21) in communication with said source; and
an analytical pan balance (20) with at least its balance pan in said cavity;
wherein the walls of said cavity form a polyhedron that focuses microwave energy of the predetermined frequency on said balance pan while supporting a plurality of TM and TE modes in said cavity,
characterized in that said cavity comprises a polyhedron with twelve faces, eight said faces (19, 24, 25, 29-31, 38, 39) having planes which form a regular octagon;
in that one said face (23) comprises a regular octagon face perpendicular to said eight faces;
in that a further face (27) of said polyhedron is an octagon parallel to said regular octagon face (23); and
in that two of said faces (26, 28) are both non-parallel and nonperpendicular to the other ten said faces.
2. A volatiles analyzer according to claim 1 wherein all sides of said polyhedron cavity are planar.
3. A volatiles analyzer according to claim 1 wherein:
said eight perpendicular faces are joined to said regular octagon face;
said parallel face (27) is joined to two of said perpendicular faces (26, 28) that are respectively parallel to one another; and
said non-parallel and non-perpendicular faces are each joined to said parallel face; and
said non-parallel and non-perpendicular faces are each joined to two of said perpendicular faces.
4. A volatiles analyzer according to any preceding claim wherein said nonparallel, non-perpendicular faces (26, 28) are perforated to encourage an airflow through said cavity.
5. A volatiles analyzer according to claim 4 further comprising a fan (33) to enhance airflow.
6. A volatiles analyzer according to any preceding claim further comprising:
a waveguide (19) between said source and said cavity and in microwave communication with both of said source and said cavity; and
a port (22) from said waveguide to said cavity.
7. A volatiles analyzer according to claim 6 wherein said port (22) is the sole entry for microwaves between said waveguide (19) and said cavity (21).
8. A volatiles analyzer according to claim 4 wherein:
said waveguide (19)is a rectangular solid; and
said port (22) is a longitudinal slot positioned in one face of said waveguide and oriented neither parallel to nor perpendicular to any of the angles forming said rectangular solid waveguide.
9. A volatiles analyzer according to any preceding claim further comprising:
an infrared sensor positioned with respect to said cavity to measure the temperature of a sample on said balance pan;
means in electronic communication with said infrared sensor for moderating the microwave power in said cavity based upon the measured temperature.
10. A volatiles analyzer according to any preceding claim further comprising a processor in electronic communication with said balance for calculating the expected final weight of a sample based upon two or more measurements of the weight at discrete time intervals as the weight changes under the influence of the microwave radiation upon the sample.
11. A volatiles analyzer according to any preceding claim wherein said microwave source is a magnetron, a klystron or a solid state device.
12. A volatiles analyzer according to any preceding claim further comprising an air shield (42) in said cavity to moderate the flow of gases through the cavity.
13. A volatiles analyzer according to claim 12 wherein the air shield (42) is formed of a grid-like framework (44) which carries one or more moisture-absorbent pads (45).
1. Analysator (10) für flüchtige Substanzen, umfassend:
eine Quelle von Mikrowellenstrahlung, die selektiv mindestens eine vorgegebene Frequenz der Mikrowellenstrahlung erzeugen kann;
einen Hohlraum (21) in Kommunikation mit der Quelle; und eine analytische Schalenwaage (20), wobei zumindest ihre Waagschale sich im Hohlraum befindet; wobei die Wände des Hohlraums einen Polyeder bilden, der Mikrowellenenergie von der vorgegebenen Frequenz auf die Schalenwaage fokussiert, während mehrere TM- und TE-Modi im Hohlraum unterstützt werden,
dadurch gekennzeichnet, dass der Hohlraum einen Polyeder mit zwölf Flächen umfasst, wobei acht Flächen (19, 24, 25, 29-31, 38, 39) Ebenen haben, die ein reguläres Oktagon bilden;
dadurch, dass eine Fläche (23) eine regelmäßige Oktagonfläche senkrecht zu den acht Flächen umfasst;
dadurch, dass eine weitere Fläche (27) des Polyeders ein Oktagon parallel zu der regelmäßigen Oktagonfläche (23) ist; und
dadurch, dass zwei der Flächen (26, 28) sowohl nichtparallel als auch nichtsenkrecht zu den anderen zehn Flächen sind.
2. Analysator für flüchtige Substanzen nach Anspruch 1, wobei alle Seiten des Polyederhohlraums planar sind.
3. Analysator für flüchtige Substanzen nach Anspruch 1, wobei:
die acht senkrechten Flächen mit der regelmäßigen Oktagonfläche verbunden sind;
die parallele Fläche (27) mit zwei der senkrechten Flächen (26, 28) verbunden ist, die parallel zueinander sind; und
die nichtparallelen und nichtsenkrechten Flächen jeweils mit der parallelen Fläche verbunden sind; und
die nichtparallelen und nichtsenkrechten Flächen jeweils mit zwei der senkrechten Flächen verbunden sind.
4. Analysator für flüchtige Substanzen nach einem der vorherigen Ansprüche, wobei die nichtparallelen, nichtsenkrechten Flächen (26, 28) perforiert sind, um einen Luftstrom durch den Hohlraum zu fördern.
5. Analysator für flüchtige Substanzen nach Anspruch 4, der ferner einen Ventilator (33) zum Verstärken des Luftstroms umfasst.
6. Analysator für flüchtige Substanzen nach einem der vorherigen Ansprüche, der ferner Folgendes umfasst: einen Hohlleiter (19) zwischen der Quelle und dem Hohlraum und in Mikrowellenkommunikation mit der Quelle und dem Hohlraum; und einen Port (22) von dem Hohlleiter zum Hohlraum.
7. Analysator für flüchtige Substanzen nach Anspruch 6, wobei der Port (22) die einzige Eintrittsmöglichkeit für Mikrowellen zwischen dem Hohlleiter (19) und dem Hohlraum (21) ist.
8. Analysator für flüchtige Substanzen nach Anspruch 4, wobei:
der Hohlleiter (19) ein rechteckiger Feststoff ist; und
der Port (22) ein längslaufender Schlitz ist, der in einer Fläche des Hohlleiters angeordnet und weder parallel noch senkrecht zu den Winkeln ist, die den rechteckigen festen Hohlleiter bilden.
9. Analysator für flüchtige Substanzen nach einem der vorherigen Ansprüche, der ferner Folgendes umfasst:
einen Infrarotsensor, der bezüglich des Hohlleiters zum Messen der Temperatur einer Probe auf der Schalenwaage angeordnet ist;
Mittel in elektronischer Kommunikation mit dem Infrarotsensor zum Mildern der Mikrowellenenergie im Hohlraum auf der Grundlage der gemessenen Temperatur.
10. Analysator für flüchtige Substanzen nach einem der vorherigen Ansprüche, der einen Prozessor in elektronischer Kommunikation mit der Waage zum Berechnen des erwarteten Endgewichts einer Probe auf der Grundlage von zwei oder mehr Messungen des Gewichts zu getrennten Zeitintervallen umfasst, wenn sich das Gewicht unter dem Einflus s der Mikrowellenstrahlung auf die Probe ändert.
11. Analysator für flüchtige Substanzen nach einem der vorherigen Ansprüche, wobei die Mikrowellenquelle ein Magnetron, ein Klystron oder eine Festkörpervorrichtung ist.
12. Analysator für flüchtige Substanzen nach einem der vorherigen Ansprüche, der ferner ein Luftleitblech (42) im Hohlraum umfasst, um den Strom von Gasen durch den Hohlraum zu zügeln.
13. Analysator für flüchtige Substanzen nach Anspruch 12, wobei das Luftleitblech (42) aus einem gitterartigen Rahmen (44) gebildet ist, der ein oder mehrere feuchtigkeitsabsorbierende Matten (45) trägt.
1. Analyseur de substances volatiles (10) comprenant :
une source de rayonnement hyperfréquence qui peut sélectivement produire au moins une fréquence prédéterminée de rayonnement hyperfréquence ;
une cavité (21) en communication avec ladite source ; et
une balance analytique à plateau (20) avec au moins son plateau de balance dans ladite cavité ;
dans lequel les parois de ladite cavité forment un polyèdre qui concentre l'énergie hyperfréquence de la fréquence prédéterminée sur ledit plateau de balance tout en supportant une pluralité de modes TM et TE dans ladite cavité, caractérisé en ce que ladite cavité comprend un polyèdre à douze faces, huit desdites faces (19, 24, 25, 29-31, 38, 39) ayant des plans qui forment un octogone régulier ;
en ce que l'une desdites faces (23) comprend une face d'octogone régulier perpendiculaire auxdites huit faces ;
en ce qu'une autre face (27) dudit polyèdre est un octogone parallèle à ladite face d'octogone régulier (23) ; et
en ce que deux desdites faces (26, 28) sont toutes deux non parallèles et non perpendiculaires aux dix autres desdites faces.
2. Analyseur de substances volatiles selon la revendication 1, dans lequel tous les côtés de ladite cavité de polyèdre sont plans.
3. Analyseur de substances volatiles selon la revendication 1, dans lequel :
lesdites huit faces perpendiculaires sont assemblées à ladite face d'octogone régulier ;
ladite face parallèle (27) est assemblée à deux desdites faces perpendiculaires (26, 28) qui sont respectivement parallèles entre elles ; et
lesdites faces non parallèles et non perpendiculaires sont chacune assemblées à ladite face parallèle ; et
lesdites faces non parallèles et non perpendiculaires sont chacune assemblées à deux desdites faces perpendiculaires.
4. Analyseur de substances volatiles selon l'une quelconque des revendications précédentes, dans lequel lesdites faces non parallèles, non perpendiculaires (26, 28) sont perforées pour favoriser un écoulement d'air à travers ladite cavité.
5. Analyseur de substances volatiles selon la revendication 4, comprenant en outre un ventilateur (33) pour améliorer l'écoulement d'air.
6. Analyseur de substances volatiles selon l'une quelconque des revendications précédentes, comprenant en outre:
un guide d'ondes (19) entre ladite source et ladite cavité et en communication d'hyperfréquence à la fois avec ladite source et avec ladite cavité ; et
un orifice (22) allant dudit guide d'ondes vers ladite cavité.
7. Analyseur de substances volatiles selon la revendication 6, dans lequel ledit orifice (22) est la seule entrée pour l'hyperfréquence entre ledit guide d'ondes (19) et ladite cavité (21).
8. Analyseur de substances volatiles selon la revendication 4, dans lequel :
ledit guide d'ondes (19) est un solide rectangulaire ; et
ledit orifice (22) est une fente longitudinale positionnée dans une face dudit guide d'ondes et orientée ni parallèlement ni perpendiculairement à l'un quelconque des angles formant ledit guide d'ondes solide rectangulaire.
9. Analyseur de substances volatiles selon l'une quelconque des revendications précédentes, comprenant en outre:
un capteur infrarouge positionné par rapport à ladite cavité pour mesurer la température d'un échantillon sur ledit plateau de balance ;
des moyens en communication électronique avec ledit capteur infrarouge pour modérer la puissance hyperfréquence dans ladite cavité en fonction de la température mesurée.
10. Analyseur de substances volatiles selon l'une quelconque des revendications précédentes, comprenant en outre un processeur en communication électronique avec ladite balance pour calculer le poids final attendu d'un échantillon en fonction de deux mesures ou plus du poids à intervalles de temps distincts lorsque le poids change sous l'influence du rayonnement hyperfréquence sur l'échantillon.
11. Analyseur de substances volatiles selon l'une quelconque des revendications précédentes, dans lequel ladite source hyperfréquence est un magnétron, un klystron ou un dispositif à état solide.
12. Analyseur de substances volatiles selon l'une quelconque des revendications précédentes, comprenant en outre un dispositif de protection d'air (42) dans ladite cavité pour modérer l'écoulement de gaz à travers la cavité.
13. Analyseur de substances volatiles selon la revendication 12, dans lequel le dispositif de protection d'air (42) est formé avec un cadre en forme de grille (44) qui porte un ou plus coussinets d'absorption d'humidité (45).

REFERENCES CITED IN THE DESCRIPTION
This list of references cited by the applicant is for the reader's convenience only. It does not form part of the European patent document. Even though great care has been taken in compiling the references, errors or omissions cannot be excluded and the EPO disclaims all liability in this regard.

Patent documents cited in the description
- US
5632921
A
[0010]
- WO
9940409
A
[0011]
- US
01866
W
[0011]
- US
9921490
W
[0026]
- US
6084226
A
[0026]
- US
4438500
A
[0027]
- US
4457632
A
[0027]
- US
39782599
A
19990917
[0030]

Non-patent literature cited in the description
- DORF
The Electrical Engineering Handbook
CRC Press
19970000
[0027]